"why not use differential mode with wide windows instead of integral mode"
That's generally how I set my PHAs up.
On 2/24/2016 4:20 PM, Ken Severin wrote:
> Hi all - I'm sending this to both lists so I can get everyone's wisdom.
> I was explaining to a class about setting up gas detectors and the
> difference between differential and integral modes. The question came
> up as to why one would ever operate in intergral mode. I didn't have
> a good answer (obviously setting up really tight parameters in
> differential mode can cause problems) - why not use differential mode
> with wide windows instead of integral mode?
> Is this a holdover from earlier electronics?
> Ken in Alaska
> You are currently subscribed to sx50-users as: [log in to unmask]
> To unsubscribe send a blank email to [log in to unmask]
John J. Donovan [log in to unmask]
University of Oregon (541) 346-4632 (office)
1443 E. 13th Ave (541) 346-4655 (probe)
Eugene, OR (541) 346-6854 (FAX)
CAMCOR Web: http://camcor.uoregon.edu/
Facility Web: http://epmalab.uoregon.edu/
Cameca SX100 Schedule: http://camcor.uoregon.edu/calendar/sx100
Cameca SX50 Schedule: http://camcor.uoregon.edu/calendar/epma
FEI Quanta Schedule: http://camcor.uoregon.edu/calendar/sem
JEOL Probe Users Listserver
Moderator: Anette von der Handt, [log in to unmask],
Electron Microprobe Lab, University of Minnesota
Post a message: send your message to [log in to unmask]
Unsubscribe: send "SIGNOFF PROBEUSERS" to [log in to unmask]
On-line help and FAQ: http://probelab.geo.umn.edu/jeoluserlist.html