JEOL Probe Users Listserver
Moderator: Ellery Frahm, [log in to unmask], Electron Microprobe Lab, University of Minnesota
Post a message: send your message to [log in to unmask]
Unsubscribe: send "SIGNOFF PROBEUSERS" to [log in to unmask]
On-line help and FAQ: http://probelab.geo.umn.edu/listserver.html
I'd like to clarify my position in the question I asked yesterday.
I have no intention of getting rid of the bubble device, and I have no intention of getting rid of P10. I use the system now with a regulator, a needle valve, and the bubble meter. My needle valve is unmarked as to range, and is really too coarse for fine control over the rate - with the side effect that if you bump the needle valve knob the flow rate changes a lot. My intention is to replace this with a finer control rotameter which has the needle control valve and a visual sight gauge to let you know that it's actually set correct. Since the JEOL spec doesn't actually say what the flow rate is in l/min (or at least I can't find it) I figured I would ask if anyone knew.
Have a nice day.