Dear JEOL users:
Our JEOL 8200 is tripping its power off and giving us an error light on panel for the oil that is used to cool the lens and some electronics. I have some questions:
Does anyone know how high the oil must be in the oil reservoir for the pump sensor not to trip off the pump? (There is a float switch).
Is it common / uncommon / or rare / for the oil elvel to drop slightly over time?
Does anyone know what type of oil it is?
And, where it can be obtained?
Department of Geosciences
The University of Tulsa
800 S. Tucker Avenue
Tulsa, OK 74104
[log in to unmask]
JEOL Probe Users Listserver
Moderator: Anette von der Handt, [log in to unmask],
Electron Microprobe Lab, University of Minnesota
Post a message: send your message to [log in to unmask]
Unsubscribe: send "SIGNOFF PROBEUSERS" to [log in to unmask]
On-line help and FAQ: http://probelab.geo.umn.edu/listserver.html